As a researcher, a technician, a lab manager, or just a visitor, you should always follow safe practices in the lab space. When doing cutting edge science, use cutting edge safety to help make new, groundbreaking achievements without endangering the people making them. Find some basic guidelines in Cutting Edge Science Needs Cutting Edge Safety
Tag: Safety Notes
Related to general safety notes
Found: “Class 2” laser pointers not as advertised
Click here to view a PDF write-up of the incident.
An academic department turned over two green laser presenters labeled “Class 2″ to the Homewood Laser Safety Advocate for evaluation because one seemed “too bright.” Normally, a Class 2 laser presentation pointer should put out no more than 1 milliwatt of energy.
Both presenters were found to be putting out more than 10 times the allowable amount of energy, including energy in the invisible infrared range, which is more dangerous. (Green laser pointers are actually infrared lasers that use special optics to generate green light from the IR.) The Laser Safety Advocate tested several additional pointers from that department, finding them all in conformance with their markings. The overpowered pointers were disposed.
The overpowered pointers were actually hazardous Class 3B lasers which should not be used in an uncontrolled lecture or presentation setting. Homewood limits the power of laser pointers to Class 2; testing has shown that brighter pointers are not necessary in any lecture hall on campus. The class of a laser device is stamped on a small yellow or white sticker on the product.
These were name-brand laser pointers purchased from nonstandard sources (e.g., online auction sites); we are as yet unsure whether they were genuine branded products that are off-specification or if they were counterfeit. Please buy all laser pointers from standard JHU-approved sources such as Office Depot; unusual distribution channels are more likely to sell counterfeit or otherwise out-of-specification products. A sample of the sample laser presenter purchased from a JHU-preferred vendor measured within normal safe tolerances.
In 2013, the National Institute of Standards and Technology (NIST) found that 90% of green laser pointers and 44% of red laser pointers were out of compliance with federal safety regulations and their markings.
If you have a laser pointer that seems too bright, especially if it is green, contact the Homewood Laser Safety Advocate, Niel Leon, [email protected]. He can test your laser pointer and return it to you if it is safe to use (or help you find a source for a safe one if it’s not).
See the HSE Guidance Document on laser pointers, as well as this fact sheet(Laser pointer fact sheet v9-170725FNL), for more details.
Equipment transfer safety
Equipment being relocated must move through public corridors and outside areas; equipment being repaired or disposed is being transferred to service or disposal personnel unfamiliar with your lab and its hazards. In all cases, you are responsible for protecting others from unknown contamination. Learn more in Equipment Transfer Safety Note.
Protect your eyes from high-intensity light
Protect your vision when working with UV germicidal lamps; lasers; welding and arc lamps; or other high–energy light sources. Special goggles limit the amount of light that can reach your eyes and skin. The type and amount of protection depends on the frequency, nature, and intensity of light. Learn more in Light eye protection.
Protect your face with a face shield
When a hazard involves a lot of energy or aggressive chemicals, your face may be at risk as well as your eyes. Also, Z87.1 or Z87.1+ rated eye protection may not be adequate to protect your eyes, so additional protection might be prudent. If you could injure your face in an accident, use a face shield to protect your face – learn more in High energy facial protection.
Protect your eyes from physical hazards
Protect your eyes from chemical hazards
Chemical hazards require eye protection specifically designed for chemical hazards. Many chemicals can cause serious damage or irritation when they get into your eyes. These include, but are not limited to, acids, caustics and solvents. When working with chemical eye hazards, wear chemical splash goggles to protect your eyes – learn more in Chemical Hazard Eye Protection.
Is your plastic gas tubing an over-inflated balloon?
Many labs use compressed gases, and often we use pressure regulators to step down the 2000-3000 psi in the cylinder to the use pressure. If the regulator can produce more than about 30 psi outlet, your plastic tubing might be in danger of rupture. Read more about how to fix this without buying a new $500 regulator in How to prevent plastic tubing rupture.
Safety and extra cost don’t go together
Many people think that safety improvements for an experiment always cost extra money. This is not true–many times, appropriate improvements avoid cost while making the research inherently safer. Read about one such case that saved the Electrical and Computer Engineering department over $10,000 (and lots of class time) in Cost reduction ECE laser teaching lab.
What eye protection do I use?
There are many different types of protective eyewear available, and each one is designed to protect against a different hazard. Having the wrong type of safety eyewear can be worse than not wearing eye protection at all. Learn about the basic types in Choosing eye protection.
This Hopkins Safety Note is the start of a series on eye protection, so look for future notes covering the different types in detail.